|T O P I C R E V I E W
||Posted - 10/19/2017 : 11:51:02
For more than a year now my GMC-320 plus is no longer counting. It comes on and the software works, but counts zero. It worked well for over a year after I bought it.
Your advice will be highly appreciated.
|2 L A T E S T R E P L I E S (Newest First)
||Posted - 10/22/2017 : 07:38:46
Check the Tube voltage too. With 10M internal meter, it should read about 150V on 1000V range. With 2M internal meter, it should read about 35V on 1000V range.
||Posted - 10/22/2017 : 06:04:25
Just throwing my thoughts out there. Perhaps the GM tube is broken, damaged? I have seen them on Ebay for ~$30 US. Might try contacting GQ Electronics directly firstname.lastname@example.org to see what they think.